Acceptance and utilization of LTER data requires that:
(1) The Principal Investigator be sent a notice stating reasons for acquiring any data and a description of the publication intentions.
(2) The Principal Investigator of the data set be sent a copy of the report or manuscript prior to submission and be adequately cited in any resultant publications.
(3) A copy of any resultant publications should be sent to:
Marine Biological Laboratory
7 MBL St.
Woods Hole, MA 02543
|Dataset URLs:||METADATA: HTML, Rich Text, XML(EML compliant) |
DATA: Comma Delimited, Excel file with Metadata and data
|Dataset Title:||Year 2011, January to September, wind sensor data, 15 minute intervals, from the Ipswich Bay Yacht Club pier located in Ipswich, MA|
|Investigator 1:|| |
|Address line 1:||Ecosystems Center|
|Address line 2:||MBL|
|Address line 3:||7 MBL St|
|Associate Investigators:||Hap Garritt|
|Keywords:||PIE LTER, primary production, Massachusetts, Ipswich, climate, meteorology, wind|
|Abstract:||Wind sensor measurements (wind speed and wind direction) January to September 2011 at the Ipswich Bay Yacht Club, Ipswich, MA, 15 minute instantaneous measurement.|
|Contact:||Plum Island Ecosystems LTER Information Manager
The Ecosystems Center
Marine Biological Lab
7 MBL St
Woods Hole, MA 02543
Phone (508) 289 7485
Online URL: http://ecosystems.mbl.edu/PIE/
|DATA FILE INFORMATION:|
|Data File URL||http://ecosystems.mbl.edu/PIE/data/MON/data/MON-SO-Met15min2011a.csv|
|Data File Name||MON-SO-Met15min2011a|
|Number of Data Records||25118|
|Other Files to Reference||MON-SO-Met15min2011b|
|Availability Status||Type 1|
|Quality Control Information|
|Maintenance Description||Final data set for Year 2011|
|Log of Changes:||Version 01, 10Jun2013, metadata and data updated to comply with importation to Drupal and LTER PASTA. Used MarcrosExportEML_HTML (working)pie_excel2007.xlsm 3/14/12 12:02 PM for QA/QC to EML 2.1.0|
|RESEARCH LOCATION:||MON-SO-IBYC Data Logger Station||Site 2|
|Geographic Description||PIE YSI water quality, OTT RLS water level and wind sensor station at Ipswich Bay Yacht Club pier, Plum Island Sound, Ipswich, MA|
|Location Bounding Box|
|West Bounding Coordinate|
|East Bounding Coordinate|
|North Bounding Coordinate|
|South Bounding Coordinate|
|OR if single point location|
|Methods:||The data file represents wind speed and wind direction data from the PIE LTER weather station located at the Ipswich Bay Yacht Club, Ipswich, MA.|
Wind speed and direction
RM Young 05103 Wind monitor
Data is recorded using a YSI DCP 6200 data collection platform with Zeno datalogger and Ecowatch software . Wind speed and direction are instantaneous measurements at 15 minute intervals. YSI DCP 6200 and Ecowatch software did not have program capability for 5 sec wind measurements and calculated 15 min averages, nor capability for determining max wind speed at a specific time.
RM Young 05103, Sensor is mounted on top of 15 ft pole which is mounted on top of IBYC pier deck railing secured to a post.
Pole 15 ft
Railing 3.3 ft above deck
Deck is ~22.5 ft above sediment bottom
High tide water level ~ 16 ft
Sensor is 40.8 ft above water bottom or ~ 24 ft (7 meters) above high tide water level
Installation of RM Young 05103 Wind Sensor at IBYC, December 20, 2005
The sensor and mounting pipe were attached next to the existing RF antenna pole used for the YSI sonde radio transmission.
The mounting pipe consisted of three, five foot galvanized threaded pipes, ¾ diameter coupled together. A 1 to ¾ reducing fitting was attached to the top to accommodate the 1 diameter fitting for the sensor.
Junction box facing South, 180 degrees opposite true North (16 degree declination
from magnetic North).
Ecowatch software, new configuration file (IBYCMet.s62) set up for the sensor.
Initially set up to record wind speed and direction every 3 minutes and radiotransmit every 15 minutes.
When the sondes are reconnected in the Spring of 2006, the IBYCChl.s62 config file will need to have the MET sensors added to it. Since the DO readings are measured on 15 minute intervals, the wind will also have to be measured on that same interval, Ecowatch can not have different measurement time intervals for different sensors. Zeno datalogger with YSI set up not capable of taking measurments every 5 seconds and then averaging for 15 minutes as with Campbell dataloggers and weather stations.
On September 19, 15:15, 2011 the YSI Data Collection Platform (DCP 6200 with Zeno datalogger) was replaced with a Campbell Scientific CR1000 data logger and LoggerNet software. Measurements conducted at 5 second intervals and 15 minute averages recorded in the CR1000.
See data file MON-SO-Met15min2011b for the remainder of 2011, September 19 to December 31, 2011.
|Variable Name||Variable Description||Units||Measurement Scale||Code Information||Number Type||DateTime Format||Missing Value Code||Missing Value Code Explanation|
|Date||date and time (DD-MON-YYYY)||datetime||DD-MON-YYYY||NA||NA = not available|
|Time||time of day (hh:mm), (eastern standard time EST), 24 hour format||datetime||hh:mm||NA||NA = not available|
|WindSpeed||wind speed (m sec-1)||meterPerSecond||ratio||real||NA||NA = not available|
|WindDir||wind direction (degrees)||degree||interval||real||NA||NA = not available|